Alpha Hand & Clothing Monitor [ TYPE : NXG_HC1_A]
Alpha Hand & Clothing Monitor NXG_HC1_A manufactured by NUCLEONIX SYSTEMS primarily serve as a personnel monitoring system for checking the contamination of hands, foot & clothing of radiation worker / technicians working in Nuclear Power Plants, reactors, Radiochemical plants and other similar installations.
This instrument has been developed using state of art SBC working on WinCE. GUI & multi-lingual user interface & corresponding voice messages are facilitated by a colour TFT monitor with integrated speakers
Guidance to the user during monitoring in the form of colour graphic monitor for left / right hand / foot & status indications during monitoring & at the end of monitoring are a standard feature of this instrument.
Advanced fault diagnostic features facilitate the user to diagnose the problem easily. Also provision for faulty channel bypassing feature has been provided. USB communication for data transfer to PC.
Each Hand detector assembly shown is modular in construction facilitating easy maintenance in case of problems.
The Alpha Hand and clothing monitor shall comprise a set of detectors and an electronic unit.
|a) Hand probes:||Number of monitoring channels: 4 (Right hand 1 & 2, Left hand 1 & 2).
Detector Type: 220x140mm Zns(Ag) scintillator based detection unit.
Detection Efficiency: >20 % overall for Am-241.
Probe Construction: Each channel shall have a 220x140mm Zns(Ag) scintillation detectors. They are fitted in the instrument so that they can be detached easily formaintenance.
Sensitive Area. > 300 sq. cm.
Protection Grill: The whole detector assembly shall be protected by a suitable thin metallic grill.
Detector dimensions and performance shall conform to IEC 61098 specifications
|b) Clothing probe:||Number of monitoring channels: one
Detector Type:160x80mm Zns (Ag) scintillator.
– Detection Efficiency: > 25% overall for plastic scintillation detector assembly with Am-241.
Probe Construction: Hand-held type. It shall be placed on a holder with micro-switch assembly on the side of the monitor. On lifting the detector, the monitoring shall be started.
The electronic unit consisting of the following :
Signal processing and display unit:
The signal processing & display unit comprising of an SBC & large TFT display perform data acquisition & control. It carries out the following functions of
High Voltage, Pulse Amplifier and Data Acquisition Unit:
This unit essentially comprises of High Voltage module, 5 channel Pulse Amplifier section, FPGA based Data acquisition board and State of Art Single Board Computer unit. DC Voltages required to bias the Electronics are generated by DC-DC convertors located on this board. Signals to/from all PMTs are routed to this unit.
Switch Mode power Supplies unit:
An AC-DC converter along with switching regulators generates the necessary LV voltages of +24V, +12V & +5V respectively to power the electronic sub-systems. It is also provided with a line filter & other EMC suppression components.
Channel Pre-amplifier unit:
This unit comprises of 5 individual pre- amplifier modules used for connecting the –ve tail pulses to TTL coming from each of the scintillation detectors. HV bias to the scintillation detectors is fed through the preamplifier unit.
Counting Range :
Timing range :
Pre-settable from 1 to 99 seconds in 1 sec steps for COUNTS, Bq/cm2, CPS or CPM modes for Hand and Foot monitoring. Time constant for checking the Clothing monitoring is 5 seconds with display being refreshed every second.
HUMAN MACHINE INTERFACE:
Indications & controls:
|Mains switch||The mains switch is provided inside the cabinet of the monitor.|
|Mains indication||Red LED to indicate mains power ON is provided on the display panel.|
|Optical sensor||The monitor shall be provided with optical sensors inside the detector cavities for sensing personnel and initializing the counting automatically.|
|Test switch||Through detachable keypad, provision to check the background is given.|
|Visual alarm||Each channel color visual mimic indication is provided on the colour LCD TFT display.|
|Audio Alarm||Loud audio tone.|
|Audio Instructions||Audio instructions shall be generated for clear, contaminated, instrument fault, monitoring in progress and incomplete operation. Voice messages shall be played back in English/Marathi/Hindi/Tamil.|
|Operational Guidance||Operational guidance messages shall be displayed Before monitoring, On Clear, On Contamination and On Incomplete operations shall be generated in English.|
|Incomplete operation||Multi-lingual textual indication accompanied by audio alert are generated when counting is interrupted.|
|Clear Indication||LCD mimic indicator will be ON when all the channels are clear.|
|Counting in progress Indication||Busy indication will be shown while Counting is ON and Time left is displayed.|
|Visual display||10” colour TFT display. Given below is a partial list of the functions being carried out by the visual display.|
|Computer interface:||The monitor has a USB port for data transfer of Acquired Data / Alarm data to a connected Pen Drive..|
|Power supplies||The monitor shall have a High voltage power supply unit for the detectors and a low voltage power supply unit which supplies the DC power supplies required for the Electronic unit. It shall have a very good line voltage and load regulation for all the supplies. It shall be fitted with Mains line filters to avoid line interferences. The High voltage output shall be adjustable by handheld configurator or host PC and EHT should be displayed on the display on demand. The EHT shall adjustable from 300 V to 1500 V DC.|
|Instrument fault indication:||Fault diagnostics are carried out periodically and any failures are reported on the display like LV, HV and detector failures. Fault indications shall be cleared automatically if normal status is resumed.|
|Self diagnostics :||The monitor has built-in self diagnostics. On being powered it shall perform tests to ensure that all components and sub systems are functioning properly. It shall check for the Power supply, High Voltage Supply, Detector, Counting and measuring circuits, Alarm Systems, Display Systems and communication port. Test points shall be provided for checking the EHT voltage and for connecting external input pulse signals.|
Dimensions and Weight:
Overall Dimensions: 1500H x 600W x 800D mm Weight: <145kg.
The instrument shall be able to withstand temperature upto 50°C and relative humidity upto 90% in radiation areas.